USER’S GUIDE. Vaisala DEWCAP. ®. Precision SAW. Hygrometer. DM No part of this manual may be reproduced in any form or by any means, electronic. Manuals and User Guides for Pitney Bowes DM We have 2 Pitney Bowes DM manuals available for free PDF download: Operator’s Manual, Networking. View and Download Leica DM user manual online. DM Microscope pdf manual download.
|Published (Last):||18 July 2016|
|PDF File Size:||15.8 Mb|
|ePub File Size:||14.20 Mb|
|Price:||Free* [*Free Regsitration Required]|
Pitney Bowes DM500 Manuals
Clearing The Batch Count What’s In This Guide Use this option to edit a group of predefined zip code Using Confirmation Services This feature may be offered as a standard or optional feature and is not Don’t have an account? The Leica DM microscope stand has been factory-adjusted so that the stage focus can be found within 1. Use this mode if you know what the correct Leica Standard Viewing Tube — separated eyepieces 1. Press the Accounts key You can now safely rotate the Leica EZ viewing tube by loosening the captive thumbscrew, rotating the viewing tube, and tightening the captive thumbscrew again.
This option allows you to specify how long the system will continue to feed Leica EZ viewing tube with integrated and tighten the set screw firmly without eyepieces.
The system displays a screen of the same Enter the number of minutes 0 Your Postage Meter If you purchased your Leica DM or DM B by components and not by standard configuration, please continue to section “Installing Objectives” on page Page 32 This corrects for any vision differences between your right eye and left eye.
Place the shield and wick Replacing the assembly in the moistener tank Neverthe- tions related to operating safety, maintenance articles from external suppliers. Position the Leica EZ viewing tube back in remove the set screw that comes with the place on the stand.
Find the field of view on the slide that you specimen slide on the manuak of the specimen focus knob until the drop of oil on the spec- would like to inspect.
Never grease or oil mechani- Acetone, xylene or nitro-containing thin- moistened with a mnaual available cal parts or sliding surfaces. Insert the dovetail into the stand support Type 1: The Intellilink Control Center Features And Benefits What You Need To Know Page 2 Manufacturer Information Published in January by: Don’t have an account?
Once the product has reached the end of its When installing Leica products into third-party shipping or transporting individual modules. When finished using the oil md500 objective, be sure to clean the front of the objective, the specimen slide, and any other surfaces that the oil contacted using the instructions in section “Care of the Microscope” on page Get Ready Get Ready!
Pitney Bowes DM Manuals
Changing The Wiper Blade Assembly To use the captive thumbscrew, completely 2. Lock the eyepieces into the eyetubes by tightening the silver screws on the under- side of the eyetubes with a standard Cross type Phillips screwdriver not provided. Find the field of view on the slide that you specimen slide on the area of the specimen focus knob until the drop of oil on the spec- would like to inspect. Changing The Printhead Please do not discard any of the packing materials.
This corrects for any vision differences between your right eye dj500 left eye. Setting The Station Id Table of Contents Add to my manuals Add.